Verical placement of 2 m ESCALON, the vertical axis of the machine has only 1000 mm stroke
Data protection procedures:
We do not monitor or evaluate or store the personal data of visitors of our web pages. Direct communications e.g. by e-mail, whatsapp and in paper form with our clients or potential clients or providers are stored as needed for the preparation of offers, contracts, manufacturing of products, elaboration of doumentation and invoicing. Such data we store confidentially for an indefinite period of time. A client or provider may contact us via the above e-mail address and request a partial or complete deletion of such data. Should you by error receive an e-mail from us which was not meant for you, please delete it and inform us.
Websites of representatives and others:
2.7 m long ESCALON Ball Beam on an 11 m x 2 m x 1 m CMM , in the 11 m long axis four stitched positions were meaasured
Note that the precision of shifting plays no first order role for the results.
2 Photographs below: Courtesy Bombardier Germany
2 m ESCALON on 3 m CMM requires stitching/overlapping in spacediagonals
Mounting a 2 m ESCALON in vertical position
The following photographs show precision-joint ESCALON bars
Courtesy: Zeiss Mexico
2.1 m beam with precision joint in middle, on mounting tube and with column fixtures of carbon fibre
The bar can be rotated in holder; this facilitates measurement with offset probe stylus according to ISOO10360-2
The joint is usually made of a ceramic ball and a ceramic ring, both precision-lapped to each other. Strong magnets hold the two bars together with a constant force
The precision joint repeats to about 0.0005 mm.
The length across the joint is calibrated.
The columns and pivot heads allow the placement in virtually any desired measurement line in the CMM volume